Nucleic Acid Purification System 32, like KingFisher 32 flex-
Q & A
Q: What is the runtime to extract 96 samples?
A: It depends on extraction kits, different extraction kits runtime different, usually 25-35 mins
Q:What kind of extraction kits can be used?
A: This machine is open source, you can use any Brand extraction kits, we recommend use large company’s extraction kits
Q:Can we use the extraction kits made by Thermo Fisher?
A: Yes, you can use Thermo and other Brand extraction kits
Q: I also need a copy of the protocols.
A: We do not sell extraction kits, You need get protocols from extraction kits company
Q:What kind of consumables do we need? Please send me the list of all the consumables needed in order to use your equipment including tip comb. I would appreciate it if I can get an item number from Thermo Fisher?
A:There are three consumables for this 96 system.
1) Part# AS-17061-01 Magnetic tip comb
2) Part# AS-17061-02 96-Deep well plate
3) Part# AS-17061-03 96-Elution plate
The usage ratio is 4:1:1, 4 96-Deep well plate : 1 Magnetic tip comb : 1 96-Elution plate
You also can use Thermo consumables, the Thermo part number is 95040460 (96-Deep well plate, sterile), 95040450 (96-Deep well plate, non-sterile), 97002534 (Magnetic tip comb )
Q:If I place the order today, can you mail that out to me today?
A: The lead time is 2 weeks after receive the payment
Q: Does this instrumentation include a heating block for any steps that require heating?
A: This machine has 8 plate positions, the No.2 and No.8 plate position have heating function
Q: Does it have a shaker for any shaking steps?
A: The Plate position is firm, cannot moving or sharking. But the program has mix function.
Q: What software does it use to program the protocols?
A: Customer can make their own protocol and save to machine. We also have our own reagent and protocol. We can build in if customer choose our reagent.